How is the X ray standing perspective bucky stand implemented?
The X ray bucky stand is provided with a sliding slot, and a pair of spring beads for bucky lifting adjustment are symmetrically embedded in the center of the upper section of the sliding slot; the plexiglass is first fixed by screws on the solid surface of the X ray bucky stand.
The partition plate is then fixed with a certain degree of hard foam block on both sides thereof to form a trapezoidal body fixing groove, and a transparent polyester film is covered with the groove shape, through the pulling rope and the elastic force.
The loose belt is provided with a fixed position plate; the other side of the X ray bucky stand is provided with a slide box, and a lower slot of the card is provided at the lower edge of the bottom edge; the bottom is welded with a spiral steel column and a column angle iron pointer, and the spiral The steel column is equipped with a rotating jacket, the top of the rotating jacket is welded with a T-shaped steel plate, the lower end is provided with an angle disc with four 90-degree scales, the lower section is provided with a fixed knob; then the X ray bucky stand is fixed to the T-shaped shape at the top of the rotating jacket.
On the steel plate, the upper plate is fixed on the T-shaped steel plate at the center of the bottom of the X ray bucky stand.