Which bucky stand should be selected for mobile DR X-ray machine
Recently, Mr. W, a medical device company, asked about the bucky stand for mobile DR. at present, the hospital has a mobile DR. he wants to install a bucky stand. The flat panel detector is wireless, with a size of 17 * 17 inches. When asked about the thickness of the flat panel detector, the customer is not sure. It is suggested that the customer and the hospital should determine the thickness of the bucky stand, because there are several types of cassette of Sinovel image. The thickness of the card slot is made below Let’s talk about it in detail.
The thickness of the slot in the film box of the positive bucky stand is 2cm, which means that the thickness of the board is not more than 2cm; the thickness of the slot in the box of the side loading bucky stand is 2.8cm, and the thickness of the board is not more than 2.8cm; the thickness of the slot in the box of the DR frame is 3cm, and the thickness of the board is not more than 3cm.
Among them, the positive bucky stand is suitable for the cassette, Cr IP board and wireless flat panel detector, the side loading frame and DR frame box are suitable for the cassette, CR IP board and wireless or wireless flat panel detector. The filter grid can be installed inside the door of the positive bucky stand and the side bucky stand, but the DR cassette can not be installed with the filter grid. We can see the difference among the three products More clearly.
As for the fixing method of the base, there are those that are directly fixed on the ground with expansion screws. We call them fixed bucky stand. If they are equipped with mobile bases, we call them mobile bucky stand. This is customized according to the needs of different users.
Our company specializes in producing all kinds of vertical and mobile chest X-ray bucky stands. If you have any questions, please contact us by email.